You can also use the PDF file to print or send to someone else
Congratulations on your new health gate.
Please do not forget to dispose of the packaging material in an environmentally friendly manner or to hand it in at a recycling station.
Your delivery includes:
1 device without base (colors may vary).
1 base and 4 screws (screws are on the inside of the door)
2 keys for the door on the back (inside of the door)
1 empty 5 liter container for the disinfectant liquid (inside of the door)
1 charger input 220 to 230 V output 5 V 3.0 A (inside of the door)
How to make the device ready for use:
(1) First, place the device horizontally in a safe place.
(2) Fasten the base plate with the included 4 screws.
(3) Put the Healthgate back in its upright position.
The on / off switch is located on the back of the device.
- Click once to switch it on (4 green LEDs full charge, one LED / 25% please recharge only with the charger included)
- Device ready for temperature measurement and hand disinfection (white LEDs flashing one after the other and red light bar below, see front display)
Automatic temperature measurement (forehead in front of the sensor) green light bar or red if outside the tolerance)
Hold hands in large opening and automatically receive disinfection mist. Rub the liquid with your hands.
- a double click to switch off (standby for temperature measurement WITHOUT disinfection, see front display with white LEDs flashing one after another).
- Temperature measurements, for example with the wrist, are always possible if there are differences in size, such as children or wheelchair users. Always keep a distance to the measuring sensor.
A 100% charge takes 5 hours. When fully charged, the device has energy for 48 hours of operation. When the device is not in use it is automatically in standby mode. The device can remain in standby mode for 72 hours.
Exchange or refilling of the storage tanks.
We recommend a complete exchange of the storage tank.
When exchanging, please dispose of the old tank in an environmentally friendly manner
The storage tank with sealing cap (4cm diameter) holds up to 5 liters, but smaller units are also possible as long as the hose is covered with liquid. Use only approved disinfectants and carefully read the instructions and warnings for each product. PLEASE DO NOT USE GEL!
The black tube with screw cap is located on the left inside of the door. After filling, close the container and place it on the bottom of the device. Then close the door with the key provided (protection)
After every exchange of the storage tank or refilling, please use the hand disinfection several times to ensure that the hose is filled.
2 battery systems.
The 1st (main) battery system also supplies the battery of the temperature measuring device (measurements even when the main battery is empty) for 48 hours
Do not leave the charger connected all the time.
Ready for use in standby mode for 72 hours.
In normal operation XXX users per hour XX hours. Please wait for the test phase.
When the batteries are charged and there is disinfectant in the reservoir, the device is ready for use.
How is the temperature measuring device used?
Keep your forehead less than 10 cm from the infrared display. The harmless blue light is a distance signal that indicates when someone is using the device. It switches back to standby mode until it is used again. Then the red infrared light reports your body temperature.
For children and wheelchair users, a measurement can also be made with the inside of the wrist.
If your body temperature is between 32.0 ° and 43.0 °, the result will be shown on the display. At a temperature below 32.0 ° an "L" for low and at a temperature above 43.0 ° an "H" for high is displayed (see front display)
If the temperature is OK, it lights up green, if the temperature is too high it lights up red.
Use with the app:
The Health Gate app can be downloaded free of charge from the App Store. Search for Health-Gate in the App Store.
Once downloaded and installed, you need to connect the app to the meter by pressing CONNECT in the top right; make sure that the measuring device is switched on beforehand.
In addition, an alarm can be set (top left), which is triggered when a measured value is higher than the set tolerance.
The maximum distance between the app and the device depends on the environment in which the measuring device is set up.
An app can be connected to several measuring devices and display the results of all recorded measurements. The temperature measuring device can only be connected to a single app (initial connection) (data protection security)
Maintain the measuring device with normal household cleaners.
For indoor use only.
Always place the device on a level surface to prevent it from tipping over.
Do not place the device in the blazing sun, but only at temperatures between 10 ° and 30 °
Only use approved disinfectants, including those for allergy sufferers.
Do not place the device near an open flame.
Always keep the door to the interior of the device closed.
Check the disinfectant fluid regularly.
Use the disinfectant on your hands only.
There is no liability for the disinfectant liquids and in connection with any unusual use of the measuring device.
The temperature reading results are for your information only and do not guarantee you are foursome. If you have any doubts, ask your doctor or pharmacist.